|K18200 Water Spray Apparatus|
Evaluates the ability of a lubricating grease to adhere to a metal surface when subjected to a direct water spray under controlled conditions. The percentage of grease sprayed off a stainless steel test panel after a specified period is determined by weight.
Features and Benefits
Conforms to ASTM D4049 specifications
Improved spray chamber design
Click on the image to view a larger photo of the instrument.
|K18200 Water Spray Apparatus
Complete Water Spray Apparatus meeting ASTM specifications, including spray chamber, delivery system and constant temperature reservoir. Sprays water at the specified rate and temperature on a test panel coated with sample grease. To test for water spray resistance, fill reservoir with 8L of tap water and set thermostat at test temperature. Circulate the water through the system to attain temperature equilibrium and insert the coated test panel in the spray chamber. Adjust water spray to 40psi (276kPa) and continue for 5 minutes. Water spray system includes 1/3hp positive displacement pump; spray nozzle with snubber fitting: 0-60psi pressure gauge; bypass valve; shut-off and drain valves; and flexible high pressure water lines. Hinged acrylic spray chamber cover is recessed into the chamber housing to insure watertight operation. Two thermometer wells permit separate monitoring of reservoir and water spray temperatures. Standardized grease application fixture coats test panel with the required thickness of sample grease. Uses tap water; does not require water hook-up.
Conforms to: ASTM D4049
Drive Motor: 1/3hp, 1725rpm
Pump: rotary gear positive displacement type
Pressure Gauge: 0-60psi
Temperature Control Stability: ±1°F (±0.5°C)
115V 60Hz, Single Phase, 13.3A
220-240V 50Hz, Single Phase, 6.8A
220-240V 60Hz, Single Phase, 6.8A
Stainless Steel Test Panel
Grease Application Fixture
Net Weight: 110 lbs (49.9kg)
Shipping Weight: 180 lbs (81.6kg)
Dimensions: 14.2 Cu. ft.
|When inquiring about NIST traceable certified thermometers, please refer to the catalog number for the corresponding thermometer and replace the middle three zeros in the catalog number with 004. Example: 251-000-01C ASTM 1C Thermometer would be 251-004-01C ASTM 1C Certified Thermometer.|